- Vatech 3D Viewers
- Hardware Sensors
- Hardware Capture Program
- Legacy Devices
- General Q&A
Please use the following search box to quickly locate valuable FAQ support guide.You can use key words or full statement to search by. Alternatively you can simply browse by Category below
Error in program callers code / CVDACQ_ETERR program Code is occurred when communication between PC (Capture program) and One Shot* Ceph sensor (OS PRO) or (OS PRO TROY) style sensor is not established, or sensor not ready state in IP communication occurs.
First you will want to turn off the X-ray device for 20 seconds and allow the unit to reset.
Next reboot the PC as well to be on a fresh boot.
Turn on the PC first and allow windows to load.
Turn on X-ray unit and allow to fully initialize fully.
Restart the capture sequence and confirm it is now able to take images again.
For IT/Certified Technicians you can check the PC's NIC (Or secondary NIC when available) IP setting should be 192.168.1.xx (Typically .88 preferred).
Ceph sensor IP will most likely be 192.168.1.80 , you can continuously ping the IP and verify if you get a response.
-If it responds; turn off the Xray unit and confirm the request times out; if it does not then something on the network is using this IP and will need to be stopped before the unit will work again.
-If it does not respond; check connections; IP settings and NIC.
If all fails contact Vatech Support.